Adelaide Research and Scholarship
Schools and Disciplines
School of Medicine
Anaesthesia and Intensive Care
Anaesthesia and Intensive Care Publications
Please use this identifier to cite or link to this item:
|Type: ||Journal article|
|Title: ||A randomized controlled trial comparing the AccuVein AV300 device to standard insertion technique for intravenous cannulation of anesthetized children|
|Author: ||Kaddoum, Roland N.|
Anghelescu, Doralina L.
Wright, Becky B.
Burgoyne, Laura Lee-Anne
|Citation: ||Paediatric Anaesthesia, 2012; 22(9):884-889|
|Publisher: ||Blackwell Publishing Ltd|
|Issue Date: ||2012|
|School/Discipline: ||School of Medicine : Anaesthesia and Intensive Care|
|Roland N. Kaddoum, Doralina L. Anghelescu, Mary E. Parish, Becky B. Wright, Luis Trujillo, Jianrong Wu, Yanan Wu, Laura L. Burgoyne|
|Abstract: ||Objectives and Aims: To evaluate the efficacy of the AccuVein AV300 device in improving the first-time success rate of intravenous cannulation of anesthetized pediatric patients. Background: The AccuVein AV300 device was developed to assist venepuncture and intravenous cannulation by enhancing the visibility of superficial veins. It uses infrared light to highlight hemoglobin so that blood vessels are darkly delineated against a red background. Methods/Materials: Patients were randomized to cannulation with the AccuVein AV300 device or standard insertion by experienced pediatric anesthesiologists. An observer recorded the number of skin punctures and cannulation attempts required, and the time between tourniquet application and successful cannulation or four skin punctures, whichever came first. Results and Conclusions: There were 146 patients with a median age of 4.6 years (range, 0.18–17.1 years), 46.6% were male, 80.8% were light skin colored, and 15.7% were younger than 2 years. The first-attempt success rates were 75% (95% CI, 63.8–84.2%) using AV300 and 73% (95% CI, 61.9–81.9%) using the standard method (P = 0.85). Patients with dark or medium skin color were 0.38 times less likely to have a successful first attempt than patients with light skin color. The difference between the two treatment groups in number of skin punctures and the time to insertion was not significant. Although the AV300 was easy to use and improved visualization of the veins, we found no evidence that it was superior to the standard method of intravenous cannulation in unselected pediatric patients under anesthesia.|
|Keywords: ||anesthesia; pediatric; intravenouscannulation; equipment|
|Rights: ||© 2012 Blackwell Publishing Ltd.|
|Appears in Collections:||Anaesthesia and Intensive Care Publications|
|View citing articles in: ||Google Scholar|
There are no files associated with this item.
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.